Risk of Death Following Application of Paclitaxel‐Coated Balloons and Stents in the Femoropopliteal Artery
|Several randomized controlled trials (RCTs) have already shown that paclitaxel‐coated balloons and stents significantly reduce the rates of vessel restenosis and target lesion revascularization after lower extremity interventions.|
Methods and Results
A systematic review and meta‐analysis of RCTs investigating paclitaxel‐coated devices in the femoral and/or popliteal arteries was performed. The primary safety measure was all‐cause patient death. Risk ratios and risk differences were pooled with a random effects model. In all, 28 RCTs with 4663 patients (89% intermittent claudication) were analyzed. All‐cause patient death at 1 year (28 RCTs with 4432 cases) was similar between paclitaxel‐coated devices and control arms (2.3% versus 2.3% crude risk of death; risk ratio, 1.08; 95% CI, 0.72–1.61). All‐cause death at 2 years (12 RCTs with 2316 cases) was significantly increased in the case of paclitaxel versus control (7.2% versus 3.8% crude risk of death; risk ratio, 1.68; 95% CI, 1.15–2.47; —number‐needed‐to‐harm, 29 patients [95% CI, 19–59]). All‐cause death up to 5 years (3 RCTs with 863 cases) increased further in the case of paclitaxel (14.7% versus 8.1% crude risk of death; risk ratio, 1.93; 95% CI, 1.27–2.93; —number‐needed‐to‐harm, 14 patients [95% CI, 9–32]). Meta‐regression showed a significant relationship between exposure to paclitaxel (dose‐time product) and absolute risk of death (0.4±0.1% excess risk of death per paclitaxel mg‐year; P<0.001). Trial sequential analysis excluded false‐positive findings with 99% certainty (2‐sided α, 1.0%).
There is increased risk of death following application of paclitaxel‐coated balloons and stents in the femoropopliteal artery of the lower limbs. Further investigations are urgently warranted.
Katsanos K, Spiliopoulos S, Kitrou P, Krokidis M, Karnabatidis D
|https://doi.org/10.1161/JAHA.118.011245, Journal of the American Heart Association. 2018;7, 6 Dec 2018|
|This is a post of a scientific or business information. The information given here is checked thoroughly by “Implant-Register”. However we can´t be responsible for the content. Contact the publisher, if you have questions. You may inform us about changes of the information to improve the Register.|
let us know
Find NEWS and PUBLICATIONS here according to your interests or use the search box.