|Many patients with certain pacemaker will have a call-back||anonymous|
|Patients with Medtronic pacemaker Adapta, Versa, Sensia, Relia, Attesta and Sphera and Vitatron series A, E, G and Q have a call-back. The affected devices have been sold in between 10.03.2017 and 07.01.2019. Out of 156.957 devices there were records from 2 patients about 4 disfunctions.|
In a certain modus the pacemaker may pause stimulating. The problem may be solved by programming.
Read the Medtronic call-back here.
|ärzteblatt.de, 14 Feb 2019|
|This is a post of a scientific or business information. The information given here is checked thoroughly by “Implant-Register”. However we can´t be responsible for the content. Contact the publisher, if you have questions. You may inform us about changes of the information to improve the Register.|
let us know